Care and maintenance instructions for the Ultrapen PT4 Free Chlorine and Temperature pen.
I. Routine Maintenance
1. ALWAYS rinse the FCE sensor with clean water after each use.
2. ALWAYS replace the soaker cap half filled with Sensor Storage Solution to prevent the
sensor from drying out after each use.
3. Do not drop, throw, or otherwise strike the PT4. This voids the warranty.
4. Do not store the PT4 in a location where the ambient temperatures exceed its specified Operating/Storage Temperature limits.
II. Battery Replacement The PT4 display has a battery indicator that depicts the life
remaining in the battery. When the indicator icon is at 3 bars, the battery is full. When the indicator icon falls to 1 bar, replace the battery with an N type battery.
Align groove in battery housing with guide bump in pen case.
1. In a clean/dry environment,
unscrew the pen cap in a counter-clockwise motion.
2. Slide the cap and battery housing out of the PT4.
3. Remove the depleted battery out of its housing.
4. Insert a new battery into the battery housing oriented with the negative end touching the spring.
5. Align the groove along the battery housing with the guide bump inside the PT4
case and slide the battery housing back in.
6. Screw the PT4 cap back on in a clockwise direction. Do not over tighten.
III. Sensor Cleaning
Cleaning the sensor:
Clean your sensor every two weeks, however this depends on application and frequency of use. Indications of a dirty sensor are slower and/or erroneous readings.
There are three critical components in your PT4 sensor; a very sensitive glass pH sensor bulb, a platinum ORP electrode, and a temperature sensor encapsulated in a small glass noid. Use extreme caution when cleaning your PT4 sensor.
To clean your sensor, select one of the following methods:
• Basic Cleaning: Using a solution made of dish soap mixed with water and a cotton swab, gently clean the inside of the sensor body and platinum electrode, rinse thoroughly with clean water, then recondition the sensor.
• Cleaning the pH Sensor Bulb: If the sensor becomes dirty, clean the sensor surface with an isopropyl soaked cotton swab. Then rinse thoroughly with clean water.
• Deep cleaning the platinum ORP electrode: Using the ORP electrode cleaning paper and water, gently clean the platinum electrode, rinse thoroughly then recondition the sensor.
To recondition the sensor: Rinse the sensor thoroughly with clean water, then allow it to soak in Storage Solution for a minimum of 1 hour (for best results allow the sensor to soak in Storage Solution overnight).
A. Calibration preparation
For maximum accuracy, fill 2 clean containers with each pH Buffer and/or ORP Standard Solution. Arrange them in such a way that you can clearly remember which is the rinse solution and which is the calibration standard/buffer. If you don’t have enough standard/ buffer, you can use 1 container of each standard/buffer for calibration and 1 container of clean water for all rinsing. Always rinse the FCE sensor between standard/buffer solutions. Ensure the FCE sensor is clean and free of debris.
B. pH Calibration using pH 7, 4, and 10 Buffer Solutions.
NOTE: You should always calibrate with pH 7 first.
1. Thoroughly rinse the PT4 by submerging the sensor in pH 7 Buffer rinse solution and swirling it around.
2. Push and release the push button to turn the PT4 on.
3. Push and hold the push button. The display will alternate between “CAL”, “FAC CAL”, “ºCºF TEMP”, “ModE SEL”, “PAr SEL”, “SOL ck”, and “ESC”.
4. Release the button when “CAL” displays.
5. The display will alternate between “PUSHnHLD” and “CAL.
6. Push and hold the button, The display will alternate between “PH” and “ORP”.
7. Release the button when “PH” is displayed.
8. The display will indicate “CAL” and the LED will flash rapidly.
9. While the LED flashes rapidly, dip the PT4 in pH 7 Buffer Calibration Solution so that the sensor is completely submerged.
10. While the LED flashes slowly, the pH calibration point will display along with “CAL”.
Swirl the PT4 around to remove bubbles, keeping the sensor submerged.
11. If the pH 7 calibration is successful, the display will indicate “SAVEd”, then “PUSHCONT” will be displayed (“PUSHCONT” will NOT be displayed if only calibrated with pH 4 or 10).
12. Push and release to continue or let the unit time out to exit after a 1-point or 2-point calibration.
13. Repeat steps 9 through 12 with pH 4 and 10 Buffer Solutions. After the 3rd calibration point is successfully saved, the display will indicate “SAVEd” and power off.
14. Verify calibration by retesting the calibration solution in solution check mode “SOL ck”, see section V below.
C. ORP Calibration using 80mV Quinhydrone, 260mV Quinhydrone, or 470mV MLC Light’s ORp Standard Solution.
NOTE: The PT4 has automatic temperature compensation in ORP calibration mode (from 15ºC to 30ºC).
1. Follow pH calibration steps 1 through 6, using ORP Solutions.
2. Release the button when “ORP” is displayed.
3. The display will indicate “CAL” and the LED will flash rapidly.
4. While the LED flashes rapidly, dip the PT4 in ORP Standard Solution so that the
sensor is completelysubmerged.
5. While the LED flashes slowly, the ORP calibration point will display along with “CAL”.
Swirl the PT4 around to remove any air bubbles, keeping the sensor submerged.
6. If the ORP calibration is successful, the display will indicate “CAL SAVEd”, then time out.
7. Verify calibration by retesting the calibration solution in solution check mode.
Save 10% on the Myron L Ultrapen PT4 Free Chlorine and Temperature Pen at MyronLMeters.com.