### Conductivity and Conductivity Meters – MyronLMeters.com

Posted by 23 Sep, 2012

TweetThe conductivity (or specific conductance) of a solution is a measure of its ability to conduct electricity. The standard unit of conductivity is siemens per meter (S/m). Conductivity measurements are used routinely in many industrial and environmental applications as a fast, inexpensive and reliable way of measuring ionic content in a solution. For example, the […]

The conductivity (or specific conductance) of a solution is a measure of its ability to conduct electricity. The standard unit of conductivity is siemens per meter (S/m).

Conductivity measurements are used routinely in many industrial and environmental applications as a fast, inexpensive and reliable way of measuring ionic content in a solution. For example, the measurement of product conductivity is a typical way to monitor and continuously trend the performance of water purification systems.

In many cases, conductivity is linked directly to the total dissolved solids (TDS). High quality deionized water has a conductivity of about 5.5 μS/m, typical drinking water in the range of 5-50 mS/m, while sea water about 5 S/m (i.e., sea water’s conductivity is one million times higher than deionized water).

Conductivity is traditionally determined by measuring the AC resistance of the solution between two electrodes.

Resistivity of pure water (in MΩ-cm) as a function of temperature

The standard unit of conductivity is S/m and usually refers to 25 °C (standard temperature). Often encountered in industry is the traditional unit of μS/cm. 106 μS/cm = 103 mS/cm = 1 S/cm. The numbers in μS/cm are higher than those in μS/m by a factor of 100 (i.e., 1 μS/cm = 100 μS/m). Occasionally a unit of “EC” (electrical conductivity) is found on scales of instruments: 1 EC = 1 μS/cm. Sometimes encountered is a so-called mho (reciprocal of ohm): 1 mho/m = 1 S/m. Historically, mhos antedate Siemens by many decades; good vacuum-tube testers, for instance, gave transconductance readings in micromhos.

The commonly used standard cell has a width of 1 cm, and thus for very pure water in equilibrium with air would have a resistance of about 106 ohm, known as a megohm. Ultra-pure water could achieve 18 megohms or more. Thus in the past megohm-cm was used, sometimes abbreviated to “megohm”. Sometimes conductivity is given just in “microSiemens” (omitting the distance term in the unit). While this is an error, it’s usually assumed to be equal to the traditional μS/cm. The typical conversion of conductivity to the total dissolved solids is done assuming that the solid is sodium chloride: 1 μS/cm is then an equivalent of about 0.6 mg of NaCl per kg of water.

A conductivity meter and probe

The electrical conductivity of a solution is measured by determining the resistance of the solution between two flat or cylindrical electrodes separated by a fixed distance. An alternating voltage is used in order to avoid electrolysis. The resistance is measured by a conductivity meter. Typical frequencies used are in the range 1–3 kHz. The dependence on the frequency is usually small, but may become appreciable at very high frequencies, an effect known as the Debye–Falkenhagen effect.

A wide variety of instrumentation is commercially available. There are two types of cell, the classical type with flat or cylindrical electrodes and a second type based on induction. Many commercial systems, Myron L meters, e.g.,  offer automatic temperature correction.

MyronLMeters.com offers many reliable conductivity meters – some analog, some digital, some pen-style, some multiparameter – but all accurate, reliable, and easy-to-use.

Analog Handheld conductivity meter

512M5: 0-5000 micromhos/microsiemens

Instant and accurate Conductivity tests

Electronic Internal Standard for easy field calibration

Fast Auto Temperature Compensation

Rugged design for years of trouble-free testing

Simple to use

Digital Handheld Conductivity, TDS, Salinity Pen

ULTRAPEN PT1 Conductivity – TDS – Salinity Pen

Accuracy of +/-1% of READING (+/-.2% at Calibration Point)

Reliable Repeatable Results

Solution modes: KCl, NaCl and 442

Automatic Temperature Compensation

Autoranging

Durable, Fully Potted Circuitry

Waterproof

Digital Handheld Multi-Parameter meter: Conductivity, TDS, Resistivity, pH, ORP, Temperature, Free Chlorine (FCE)
Memory Storage: Save up to 100 samples w/ Date & Time stamp
Waterproof

Digital In-Line Conductivity Monitor/Controller

The unique circuitry of our 750 Series II Conductivity Inline Meters guarantees accurate and reliable measurements. Drift-free performance is assured by “field proven” electronics, including automatic DC offset compensation and highly accurate drive voltage.

Since Temperature Compensation is at the heart of accurate water measurement, all Myron L Monitor/controllers feature a highly refined and precise TC circuit. This feature perfectly matches the water temperature coefficient as it changes. All models are corrected to 25’C. The TC may be disabled to conform to USP requirements.

Comments Off on Conductivity and Conductivity Meters – MyronLMeters.com Categories : Product Updates, Technical Tips

### Myron L Meters Thanks Dalscorp of Brunswick, Maine!

Posted by 3 Jul, 2012

Tweet Myron L Meters thanks Dalscorp engineering consultants of Brunswick, Maine, proud owners of a new Ultrapen PT1 conductivity/TDS/salinity pen. ULTRAPEN PT1 Conductivity/TDS/Salinity Pen. This instrument is designed to be extremely accurate, fast and simple to use in diverse water quality applications. Advanced features include the ability to select from 3 different solution types that […]

Myron L Meters thanks Dalscorp engineering consultants of Brunswick, Maine, proud owners of a new Ultrapen PT1 conductivity/TDS/salinity pen.

 ULTRAPEN PT1 Conductivity/TDS/Salinity Pen. This instrument is designed to be extremely accurate, fast and simple to use in diverse water quality applications. Advanced features include the ability to select from 3 different solution types that model the characteristics of the most commonly encountered types of water; proprietary temperature compensation and TDS conversion algorithms; highly stable microprocessor-based circuitry; user-intuitive design; and waterproof housing. A true, one-handed instrument, the PT1 is easy to calibrate and easy to use. To take a measurement, you simply press a button then dip the pen in solution. Results display in seconds.       ULTRAPEN – PT1    1. Push Button – turns instrument on in saved measurement mode; selects mode and unit preferences. 2. Pen Cap – provides access to battery for replacement. 3. Clip – holds pen to shirt pocket for secure storage. 4. Battery Indicator – indicates charge left in battery. 5. Display – displays measurements, mode options and battery indicator. 6. LED Indicator Light – indicates when to dip instrument in solution, when measurement is in progress, and when to remove instrument from solution. 7. Electrodes – measure electric current of solution. 8. Cell – contains flux field in defined area for accurate current measurement. 9. Scoop – contains sample solution for measurement when sampling from a vertical stream. To use, slide the open end of the scoop over the bottom of the pen until the neck of the scoop is flush with the top of the cell. Hold pen with scoop end under stream. Rinse and fill with sample solution 3 times. Fill with solution again, then take measurement. We recommend you recalibrate the pen using the scoop to retain accuracy of ±1%.
Comments Off on Myron L Meters Thanks Dalscorp of Brunswick, Maine! Categories : MyronLMeters.com Valued Customers

### MyronLMeters.com Announces the Myron L Ultrapen PT1

Posted by 28 Feb, 2011

TweetMyronLMeters.com today announced the imminent release of a new Myron L product, the Myron L Ultrapen PT1, a reliable, easy-to-use pen-style meter that measures 3 parameters – conductivity, TDS, and salinity. FOR IMMEDIATE RELEASE San Diego, California, United States of America(Free-Press-Release.com) February 27, 2011 — MyronLMeters.com today announced the arrival of a new Myron L product, the […]

### FOR IMMEDIATE RELEASE

San DiegoCaliforniaUnited States of America(Free-Press-Release.com) February 27, 2011 —

MyronLMeters.com today announced the arrival of a new Myron L product, the Myron L Ultrapen PT1, a reliable, easy-to-use pen-style meter that measures 3 parameters – conductivity, TDS, and salinity.

“The Myron L Ultrapen PT1 will be available soon

Myron L Ultrapen Conductivity TDS Salinity Meter

at MyronLMeters.com,” said James Rutan, president, “and it’s going to be a great seller. The Pocket Tester’s convenience, durability, accuracy, and reliability are unmatched in the industry. It comes complete with a battery installed (and an extra), holster, lanyard, scoop, and instructions. The Myron L PT1 is waterproof, has fully secured circuitry and, as always, MyronLMeters.com has made it easy to order. At MyronLMeters.com, you get a 10% discount…just for ordering online. In addition, all Myron L meters in stock will ship the next business day. The quality of the Ultrapen PT1 and the company’s great reputation for reliable meters is sure to make this a big hit. It even looks great!”

Myron L meters are renowned for their accuracy, reliability, and ease of use, and have applications in automatic rinse tank controls, boiler and cooling towers, circuit board cleanliness testing, deionized water, environmental applications, fountain solutions, dialysis, horticulture, hydroponics, ORP (oxidation reduction potential)/Redox, pool and spa, reverse osmosis, textiles.

MyronLMeters.com has a well-established a web presence on Facebook, Gravatar, Twitter, Squidoo, LinkedIn, and WordPress. MyronLMeters.com encourages customers to join them on these sites for special offers and discounts.

More at http://www.myronlmeters.com

Comments Off on MyronLMeters.com Announces the Myron L Ultrapen PT1 Categories : Product Updates