How long will my Standard Solutions and Buffers last?
The warranty on all standards and buffers is one year from the date it is manufactured (see the label on the bottle). If the standards and buffers become contaminated by the user pouring test samples back into the bottle or inserting the probe into the bottle the solution will not be accurate and should be discarded. The life of standards and buffers can exceed 1 year if the bottle is stored tightly capped and is not exposed to direct sunlight or freezing temperatures. If the solution becomes frozen, do not remove the cap – allow the standard or buffer solution to thaw completely and shake the bottle vigorously before opening.
How do I clean the conductivity cell cup on the handheld units?
With everyday sampling, the cell cup may build up a residue or film on the cell walls that may cause the readings to become erratic. Use a 50/50 mixture of a common household cleaner (i.e. Lime-A-Way, CLR, Tilex, etc) and DI water. Pour into conductivity cell cup and scrub with a q-tip. Be sure to get around all the electrodes and the thermistor probe. On the DS handheld unit, use an acid brush to scrub the cell cup. Let it set for about 10 minutes. Rinse the cell cup thoroughly with tap water, then a final rinse with DI water.
The display on my Ultrameter II 6P reads “Error 1”. What does that mean?
This is possibly caused by contamination to the circuit board. One or more of the traces on the PCB have been jumped/bridged and there is a contamination. Possible moisture, condensation, dirt, dried salts or other condensation inside is a potential cause for this display.
Where can I get an operations manual for my meter?
Go to MyronLMeters.com. Click on Manuals and Literature at the top of the page. Once on the Manuals and Literature page, you’ll find application bulletins, operations manuals, material safety data sheets, and product datasheets. All are free, downloadable pdf files.
How do I pick the correct range module for my Monitor or Monitor/Controller?
Pick a range module that covers 2/3 of your operating range. If you pick a range module that is too broad, then your accuracy will suffer or it will not show a number on the display. For example, if your operating range is 100-150 microsiemens, a range module of 0-200 microsiemens (-115) would be a good choice. A range module of 0- 5,000 microsiemens (-123) would not be a good choice for this application