*Note: This procedure applies to the Ultrameter, PoolPro, TechPro, and D-6 Dialysate meter.
a. Fill and rinse the conductivity cell three times with a KCL standard solution. In this example, we’re using KCL-7000.
b. Refill conductivity cell with same standard solution you rinsed with.
The “CAL” icon will appear on the display.
to step the displayed value toward the standard’s value.
In this example, we’re pressing
to go down from 7032 to 7000. You can also hold a key down to scroll rapidly.
once to confirm the new value and end the calibration.You can find technical advice and videos, the calibration solutions you need, and reliable Myron L meters at MyronLMeters.com