Most Myron L analog meters have a battery indicator glow light visible through the small hole on the lower right-hand corner of the meter face plate. If this light fails to glow when the black button is pressed, replace both batteries.
To replace the batteries detach the battery connectors. Pull on the plastic straps to remove the batteries. Replace with fresh zinc carbon or alkaline 9 volt batteries. Reinsert the plastic straps to secure batteries.
Self-conditioning of the built-in electrodes occurs each time the button is pressed with a sample in the cell cup. This ensures consistent results each time. With some samples a small downward swing of the pointer is a result of this conditioning action. This action is powerful and removes normal films of oil and dirt. However, if very dirty samples – particularly scaling types – are allowed to dry in the cell cup, a film will build up. This film reduces accuracy. When there are visible films of oil, dirt, or scale in the cell cup or on the electrodes, scrub them lightly with a small brush and household cleanser. Rinse out the cleanser, and the meter is ready for accurate measurements. pH SENSOR The unique pH electrode in your pDS meter is a nonrefillable combination type which features a porous Teflon* liquid junction (covered by U.S. Patent No. 4128468). It should never be allowed to dry out (see pH MEASUREMENT). If it does, the sensor can sometimes be renewed by soaking in a saturated potassium chloride (KCI) solution for several days. “Drifting” can be caused by a film on the sensor bulb. Use a liquid cleaner such as Windex™ or fantastik™ to clean it. The sensor buIb is very thin and delicate. Excessive pressure during cleaning may break it. Leaving high pH (alkaline) solutions in contact with the pH sensor for long periods of time can damage it. Rinsing such liquids from the pH compartment and moistening it with 4 buffer or tap water will extend its useful life. Samples containing chlorine, sulphur, or ammonia can “poison” any pH electrode. If it is necessary to measure the pH of any such sample, thoroughly rinse the pH sensor with clean water immediately after taking the measurement. Any sample element which will reduce (add an electron to) silver, such as cyanide, will attack the reference electrode. Replacement sensors are available from MyronLMeters.com. *™ DuPont Company
WATER INSIDE THE METER
Your Myron L meter is a rugged instrument and will withstand water exposure around its cell, meter movement, and switches. However, care should be taken to keep water from leaking in around the bottom cover. It is not sealed (to prevent condensation from forming). If the water is relatively clean (i.e., tap water or better), and there are only a few drops inside the meter, dry it as described below. Large amounts of water, or corrosive or very dirty solutions will almost certainly damage the meter movement or electronics. Such meters should be returned to the Myron L Company for repair.
To dry your meter:
1. Shake excess water out of the inside of the meter.
2. Dab the exposed surface dry with an absorbent cloth or tissue. Avoid pushing any water into the Calibration Controls or the switches.
3. Air dry the meter in a warm area with the bottom cover off. Allow several hours for thorough drying. If the water entered through a leak in the case or cell, or if the instrument shows erratic readings or other unusual behavior, return it for servicing.
For more on repairs and maintenance, or to download an operations manual, please visit us HERE.